(a) A DOT alcohol test must take place at an alcohol testing site meeting the requirements of this section.
(b) If you are operating an alcohol testing site, you must ensure that it meets the security requirements of § 40.223.
(c) If you are operating an alcohol testing site, you must ensure that it provides visual and aural privacy to the employee being tested, sufficient to prevent unauthorized persons from seeing or hearing test results.
(d) If you are operating an alcohol testing site, you must ensure that it has all needed personnel, materials, equipment, and facilities to provide for the collection and analysis of breath and/or saliva samples, and a suitable clean surface for writing.
(e) If an alcohol testing site fully meeting all the visual and aural privacy requirements of paragraph (c) is not readily available, this part allows a reasonable suspicion or post-accident test to be conducted at a site that partially meets these requirements. In this case, the site must afford visual and aural privacy to the employee to the greatest extent practicable.
(f) An alcohol testing site can be in a medical facility, a mobile facility (e.g., a van), a dedicated collection facility, or any other location meeting the requirements of this section.
(a) If you are a BAT, STT, or other person operating an alcohol testing site, you must prevent unauthorized personnel from entering the testing site.
(1) The only people you are to treat as authorized persons are employees being tested, BATs, STTs, and other alcohol testing site workers, DERs, employee representatives authorized by the employer (e.g., on the basis of employer policy or labor-management agreement), and DOT agency representatives.
(2) You must ensure that all persons are under the supervision of a BAT or STT at all times when permitted into the site.
(3) You may remove any person who obstructs, interferes with, or causes unnecessary delay in the testing process.
(b) As the BAT or STT, you must not allow any person other than you, the employee, or a DOT agency representative to actually witness the testing process (see §§ 40.241-40.255).
(c) If you are operating an alcohol testing site, you must ensure that when an EBT or ASD is not being used for testing, you store it in a secure place.
(d) If you are operating an alcohol testing site, you must ensure that no one other than BATs or other employees of the site have access to the site when an EBT is unsecured.
(e) As a BAT or STT, to avoid distraction that could compromise security, you are limited to conducting an alcohol test for only one employee at a time.
(1) When an EBT screening test on an employee indicates an alcohol concentration of 0.02 or higher, and the same EBT will be used for the confirmation test, you are not allowed to use the EBT for a test on another employee before completing the confirmation test on the first employee.
(2) As a BAT who will conduct both the screening and the confirmation test, you are to complete the entire screening and confirmation process on one employee before starting the screening process on another employee.
(3) You are not allowed to leave the alcohol testing site while the testing process for a given employee is in progress, except to notify a supervisor or contact a DER for assistance in the case an employee or other person who obstructs, interferes with, or unnecessarily delays the testing process.
(a) The DOT Alcohol Testing Form (ATF) must be used for every DOT alcohol test. The ATF must be a three-part carbonless manifold form. The ATF is found in Appendix G to this part. You may view this form on the ODAPC web site (http://www.transportation.gov/odapc).
(b) As an employer in the DOT alcohol testing program, you are not permitted to modify or revise the ATF except as follows:
(1) You may include other information needed for billing purposes, outside the boundaries of the form.
(2) You may use a ATF directly generated by an EBT which omits the space for affixing a separate printed result to the ATF, provided the EBT prints the result directly on the ATF.
(3) You may use an ATF that has the employer's name, address, and telephone number preprinted. In addition, a C/TPA's name, address, and telephone number may be included, to assist with negative results.
(4) You may use an ATF in which all pages are printed on white paper. You may modify the ATF by using colored paper, or have clearly discernable borders or designation statements on Copy 2 and Copy 3. When colors are used, they must be green for Copy 2 and blue for Copy 3.
(5) As a BAT or STT, you may add, on the “Remarks” line of the ATF, the name of the DOT agency under whose authority the test occurred.
(6) As a BAT or STT, you may use a ATF that has your name, address, and telephone number preprinted, but under no circumstances can your signature be preprinted.
(c) As an employer, you may use an equivalent foreign-language version of the ATF approved by ODAPC. You may use such a non-English language form only in a situation where both the employee and BAT/STT understand and can use the form in that language.
(a) No, as an employer, BAT, or STT, you are prohibited from using the ATF for non-DOT alcohol tests. You are also prohibited from using non-DOT forms for DOT alcohol tests. Doing either subjects you to enforcement action under DOT agency regulations.
(b) If the STT or BAT, either by mistake, or as the only means to conduct a test under difficult circumstances (e.g., post-accident test with insufficient time to obtain the ATF), uses a non-DOT form for a DOT test, the use of a non-DOT form does not, in and of itself, require the employer or service agent to cancel the test. However, in order for the test to be considered valid, a signed statement must be obtained from the STT or BAT in accordance with § 40.271(b) .
ASDs listed on ODAPC's Web page for “Approved Screening Devices to Measure Alcohol in Bodily Fluids” and EBTs listed on ODAPC's Web page for “Approved Evidential Breath Measurement Devices” are the only devices you are allowed to use to conduct alcohol screening tests under this part. You may use an ASD for DOT alcohol tests only if there are instructions for its use in this part. An ASD can be used only for screening tests for alcohol, and must not be used for confirmation tests.
[82 FR 52246, Nov. 13, 2017]
(a) EBTs on ODAPC's Web page for “Approved Evidential Breath Measurement Devices” that meet the requirements of paragraph (b) of this section are the only devices you may use to conduct alcohol confirmation tests under this part.
(b) To conduct a confirmation test, you must use an EBT that has the following capabilities:
(1) Provides a printed triplicate result (or three consecutive identical copies of a result) of each breath test;
(2) Assigns a unique number to each completed test, which the BAT and employee can read before each test and which is printed on each copy of the result;
(3) Prints, on each copy of the result, the manufacturer's name for the device, its serial number, and the time of the test;
(4) Distinguishes alcohol from acetone at the 0.02 alcohol concentration level;
(5) Tests an air blank; and
(6) Performs an external calibration check.
(a) As an EBT manufacturer, you must submit, for NHTSA approval, a quality assurance plan (QAP) for your EBT before ODAPC places the EBT on its Web page for “Approved Evidential Breath Measurement Devices.”
(1) Your QAP must specify the methods used to perform external calibration checks on the EBT, the tolerances within which the EBT is regarded as being in proper calibration, and the intervals at which these checks must be performed. In designating these intervals, your QAP must take into account factors like frequency of use, environmental conditions (e.g., temperature, humidity, altitude) and type of operation (e.g., stationary or mobile).
(2) Your QAP must also specify the inspection, maintenance, and calibration requirements and intervals for the EBT.
(b) As the manufacturer, you must include, with each EBT, instructions for its use and care consistent with the QAP.
(c) As the user of the EBT (e.g., employer, service agent), you must do the following:
(1) You must follow the manufacturer's instructions (see paragraph (b) of this section), including performance of external calibration checks at the intervals the instructions specify.
(2) In conducting external calibration checks, you must use only calibration devices appearing on NHTSA's CPL for “Calibrating Units for Breath Alcohol Tests.”
(3) If an EBT fails an external check of calibration, you must take the EBT out of service. You may not use the EBT again for DOT alcohol testing until it is repaired and passes an external calibration check.
(4) You must maintain records of the inspection, maintenance, and calibration of EBTs as provided in § 40.333(a)(3).
(5) You must ensure that inspection, maintenance, and calibration of the EBT are performed by its manufacturer or a maintenance representative certified either by the manufacturer or by a state health agency or other appropriate state agency.
(a) As an ASD manufacturer, you must submit, for NHTSA approval, a QAP for your ASD before NHTSA approves it and ODAPC places the device on its Web page for “Approved Screening Devices to Measure Alcohol in Bodily Fluids”. Your QAP must specify the methods used for quality control checks, temperatures at which the ASD must be stored and used, the shelf life of the device, and environmental conditions (e.g., temperature, altitude, humidity) that may affect the ASD's performance.
(b) As a manufacturer, you must include with each ASD instructions for its use and care consistent with the QAP. The instructions must include directions on the proper use of the ASD, and, where applicable the time within which the device must be read, and the manner in which the reading is made.
(c) As the user of the ADS (e.g., employer, STT), you must follow the QAP instructions.
(d) You are not permitted to use an ASD that does not pass the specified quality control checks or that has passed its expiration date.
(e) As an employer, with respect to breath ASDs, you must also follow the device use and care requirements of § 40.233 .